ComfyRay Positioning - Posterior
In this video you will learn about how to properly assemble & place the Apex Dental Sensor and holder in the patients mouth.
Difficulty: Beginner Video Time: 2:16 In this video I’m going to show you how to properly place and assemble the ComfyRay Dental Sensor Holder with the Apex Dental Sensor for an Posterior X-Ray.
More Training and Resources
- Make sure the exposure settings are correct on the X-ray source (60-70 KVP suggested). You will need to adjust exposure times (pulses), depending on the X-ray source and the physical size of the patient. Remember that maxillary molar images need the most exposure time and mandibular anterior images need the least amount of time.
- Place the sensor parallel to the teeth, but not necessarily up against the teeth.
- Maxillary shots—place the sensor at the midline of the palate.
- Mandibular shots—place the sensor more towards the tongue if necessary.
- Use a long cone (16″) technique for clearer images.
- Make sure the bite block is fully in the patient’s mouth on molar shots.
- If patient has trouble closing have them stabilize the arm of the holder with their finger.
- Make sure the cone of the X-ray source is parallel to the arm of the holder and directed at the center of the aiming ring.
- Use a cotton roll as a cushion between the sensor and tori.
- For especially sensitive patients, there are foam cushions specifically made for use with sensors. Many of these work very well with sensitive patients and make placement much more comfortable.